17 Apr 2002
The LOGR53 main CPU board (VMLOGR53) normally operates in RS232 mode.
See the LOGR53 Functional Description for
info on changing to RS485 operation as well as the RS485 note
at the end of this document.
Refer to LOGR53 CBLAST/SECNAV Commands for typical
full command descriptions.
- The following equipment should be available:
- Computer running a terminal program (ProComm Plus recommended) with RS232 output set for 9600, N, 8, 1.
Run with caps-lock ON.
- Power supply with output of 12-15 vdc and capacity of 100 ma. Make sure that the polarity
is correct otherwise the LOGR53 may be damaged!. A meter in series with the LOGR53 power
supply is recommended to monitor the supply current (less than 10 MA is typical after the
sign-on message has been sent if no MET instruments are connected).
- If the LOGR53 logger housing is open, use the RS232 adapter cable
to connect from a DB9 Male RS232
on your PC to the 4-pin Molex that terminates the standard LOGR53 power/comms cable. This adapter
includes a 2-pin Molex connector can mate to a standard LOGR53 battery or power supply adapter cable.
Otherwise, use an SAE adapter cable to connect from the console SAE bulkhead connector to the
RS232 adapter cable.
- Make sure the power supply is OFF.
- Hook up everything, start the terminal program and get it set for
the correct COM port and 9600 baud, N, 8, 1 operation.
- Power up and check for power consumption of less
than 50 MA at outset. Within a few seconds, you'll see a sign-on message, and once the "Running" message
appears (up to 15 seconds after the initial message), the current consumption typically drops to about
10 MA with the supply set for about 13.5 VDC if no ASIMET modules are attached. Idle current WITH ASIMET modules
will be around 30 MA with miscellaneous peaks during individual module internal activities, and at the top
of each minute as the LOGR53 polls each module for new data.
Individual ASIMET modules may be connected and disconnected while the LOGR53 logger is running. Each minute, as
modules are polled, missing modules are flagged, and the STAT
command will report comms failures during all but the final 15 seconds of the logging interval (see
RI setting, typically 60 seconds). For example, if
there is no response from the PRC module when the LOGR53 logger polls, the
STAT command will report "PRC failure" in its
output message for the next 45 seconds; power to the module is also turned OFF during this time, and 15
seconds before the end of the interval, it is restored. Hopefully the module recovers for this next cycle;
note that modules DO NOT produce "good" data values until they rollover their internal 60-second sampling
intervals; thus, for example, a WND module will respond the first minute after it is connected to the
LOGR53 logger (so there will be no failure message for the next minute) but will not have good data until the
next minute. This is normal; ALL ASIMET modules operate asynchronously on their individual internal
The LOGR53 (like ASIMET) has a communications protocol based on the SAIL (Serial ASCII Interface Loop)
protocol developed at Oregon State University. The module is interrogated by sending commands with the
following format (always with capital letters and terminated with a RETURN <cr>):
The STAT command will display several lines of status info about the instrument.
Note that unlike the ASIMET and VMCM2 series instruments, the LOGR53 software does not use
SAIL Addresses currently, just the preceding # character, and always a terminating
RETURN (ASCII carriage return <cr>)
On power up, the LOGR53 logger will always come up running and logging as last configured (the
STAT command will indicate Sampling GO when
the LOGR53 logger is running).
NOTE that only a subset of the commands is available while the LOGR53 logger is running.
The H command will indicate which commands are
available depending upon the GO / STOPPED state of the LOGR53 logger.
Refer to LOGR53 Commands document for full command descriptions.
Deployment Quick Start
For the complete discussion of LOGR53 system preparation for deployment, refer to the
LOGR53 System Prep document.
The important steps are:
- Check the basic operation of the system.
- Startup looks normal on the attached PC/terminal.
- After a minute (RI set to 60 seconds), D
and E commands give reasonable results from
the MET modules.
- With Sampling STOPPED, T commands give
- Set the clock if necessary (CLOCK command).
- Erase the FLASH memory card and see that the correct size is recognized.
- F commands to access FLASH card.
- UOK command to access
FEOK full erase command.
- Be sure everything returned by the STAT command looks correct - especially
Sampling GO if you are already running on the deployment batteries!!!
- The deployment batteries are NEW batteries, right?
- NOTE: If you did a complete deployment prep per the LOGR53 System Prep
document, you can wait to connect the deployment batteries until shortly before deploying the LOGR53 system;
Remember that the LOGR53 logger will ALWAYS power up RUNNING and LOGGING to FLASH memory.
- Seal up the LOGR53 logger and go....
RS232 to RS485 converters
- Black Box, Inc. sells 2 converters that work well. The first is a
PCMCIA/RS485 port (Part# IC114A, $199). This unit works well on laptops with PCMCIA slots but
does not suppress the transmissions (you see two sets of what happens).
- The second is a standalone unit (Part# IC108A, $259). This unit works well on the bench.
The settings for the IC108A standalone unit are as follows (* indicates default):
XW1A DCE *
W8 B-C 2-wire
W15 A-B * RTS/CD enabled
W5 A-B * RTS/CTS delay normal
W9 RTS/TCS delay C* 0 msec
W17 D 0.7 msec time driver remains enabled after a low-to-hi transition
W16 B * 0.1 msec Turnaround delay
S1 OUT * Normal
S2 OFF * RS-485 receiver Unterminated
S3 OFF * Line Bias Off
term 1, Rx B+ (connect to 3, Tx, B+) to the +485 (orange)
term 2, Rx A- (connect to 4, Tx, A-) to the -485 (brown)
term 3 Tx B+
term 4 Tx A-
- An RS485 adapter cable connects from a DB25 Male RS232
on the Black Box IC108A to the 4-pin Molex that terminates the standard LOGR53 power/comms cable.
This adapter includes a 2-pin Molex connector can mate to a standard LOGR53 battery or power supply