See ASIMET MODULE OPERATIONS document for instructions
on how to hook up the comms link to your PC and RS-485 adapter. This board may only be used both as a
front-end interface board in conjunction with an VOS/ASIMET C530 logger board.
Refer to Latest Version Information for reference info on latest firmware version numbers.
For further info on the operation of the Gill sensor, refer to the Gill Instruments WindObserver II Ultrasonic Anemometer user manual (available online at http://www.gill.co.uk/)
Note 1: All commands MUST be terminated with <cr> (ASCII carriage return). This is different from other ASIMET front-end boards.
Note 2: commands sent are indicated in BOLD, responses in normal text. <crlf> represents ASCII carriage return and line feed.
Firmware GILWND45 V1.x A - Address acknowledge B - Display GILL and Compass (do D,G first) C - Get Everything - single line output D[0,1] - Get/Proc Compass Direction data E - Display both GILL and TCM2 E[0,1] - Get GILL or TCM2 G[0,1] - Get/Proc GILL U,V data H - Display Help message STAT - System Status info T - Test mode UOK - Update EEPROM constants
#GIL01STAT module id from EEPROM module model name from EEPROM module serial number from EEPROM configuration date from EEPROM firmware name and version from program EPROM
#GIL01T 2.42 0.83 313.7 20.2 251.7 -0.3 0.1<crlf> 2.42 0.83 313.7 20.2 251.7 -0.3 0.1<crlf> 2.42 0.83 313.7 20.2 251.7 -0.3 0.1<crlf> 2.42 0.83 313.7 20.2 251.7 -0.3 0.1<crlf> . .
To change the current module address: ADR=GIL01<cr> To see the address: ADR<cr> To see the current setup date: CD<cr> To change it: CD=12Apr08<cr>